Custom Waveform Generation Why settle for a basic function generator when you can get arbitrary wave-forms at no extra cost? Hopefully your drivers have similar VIs. I hope someone will have an idea! With variable edge time, pulse width and voltage level, the Agilent A is ideally suited to a wide variety of pulse applications. View Cart My Account. I don’t know why but for some waveforms the Agilent starts beeping for a few minutes, and then I’ve got a long list of errors.
|Date Added:||3 April 2009|
|File Size:||5.98 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Connectivity Standard IntuiLink Software included. Don’t hesitate to ask questions if you need more details, or if I was not clear. Message 3 of 3. I too 3350 issues creating ARB waveforms. With variable edge time, pulse width and voltage level, the Agilent A is ideally suited to a wide variety of pulse applications.
So I didn’t solve my problem, still have the error with my Agilent a, but changing it to a new one made my life easier. I’m thus seeking for help. Try it and see if it helps.
Looking at your manual, my A seems to be very similar to your ARB just newer. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Not sure if they are the same issues you are having.
Agilent 33250A Function / Arbitrary Waveform Generator Instrument Driver
I also give the different subVIs that are used to configure the Agilent. Most Active Software Boards: The ARB would throw an error if the file already existed in volatile memory. You can also store up to four 64K-deep arbitrary wave-forms in non-volatile memory with user-defined names to help agiilent find the right waveform when you need it most.
Once my signal is created, I send it to the Agilent as an arbitrary waveform that is bursted once. The signal generation is OK. The included Agilent IntuiLink soft-ware allows you to easily create, edit, and download complex waveforms using the IntuiLink Arbitrary Waveform Editor. The weird part is that avilent doesn’t happen all the time but depends on the parameters I’m using for the signal.
I don’t have an Agilent but a Keysight A. I’m guessing that this issue is a format issue. The easiest way was to use the Clear Arbitrary Memory. Hopefully your drivers have similar VIs. Issue with arbitrary waveform and Agilent A.
Color Graphical Display The unique design of the A com-bines a low-profile instrument with the benefits of a color graphical display. Linear or logarithmic sweeps can be performed with a programmable frequency marker signal. I have also written a VI that generates a signal agileht I want to send to gailent Agilent as an arbitrary waveform.
IntuiLink pro-vides the tools ayilent easily create, down-load, and manage waveforms for your Agilent A. I also have many more points available per waveform. The knob or numeric keypad can be used to adjust frequency, amplitude and offset. Something might go wrong when converting the data array into binary block chain, but I don’t understand what. I had to download it first to volatile memory then transfer it to non-volatile. If I change the frequency to 8,9,10,11 or 12 kHz with all other parameters unchanged it doesn’t work.
Issue with arbitrary waveform and Agilent A – Discussion Forums – National Instruments
Message 1 of 3. The unique design of the A com-bines a low-profile instrument with the benefits of a color graphical display.
Invalid character”, and also sometimes by “error